Measuring the volume of drinking water.passing thriugh the pipeline.
1.Adopt volumetric rotarg piston principle of measurent.
2.High semsitivity .enables the measurement even at a low flow vate of 2L/h
3.Vacuum Sealed register to keep a clear ptewentation .in long tevm service.
4.Mechanical pavts use of high quality meterial to ensyuve a long life span.
5.Main casing use of stainless steel with a nice look and high sturdiness.
(The main casing of LYH-8S is use of plast)Flex Banner
Water temperature≤40℃ Water pressure≤0.6MPa
COMPLIANCE WITH STANDARD:
At low zone is±5% from minimum flow rate(qmin)to
transitional flow rate(qt)exclusive boundary
At high zone is±2% from transitional flow rate(qt) to overload flow rate(qs)